# Response factor

Last updated

Response factor, usually in chromatography and spectroscopy, is the ratio between a signal produced by an analyte, and the quantity of analyte which produces the signal. Ideally, and for easy computation, this ratio is unity (one). In real-world scenarios, this is often not the case.

## Expression

The response factor $f_{i}$ can be expressed on a molar, volume or mass  basis. Where the true amount of sample and standard are equal:

$f_{i}={\frac {A_{i}}{A_{st}}}f_{st}$ where A is the signal (e.g. peak area) and the subscript i indicates the sample and the subscript st indicates the standard.  The response factor of the standard is assigned an arbitrary factor, for example 1 or 100. Response factor of sample/Response factor of standard=RRF

## Chromatography

One of the main reasons to use response factors is to compensate for the irreproducibility of manual injections into a gas chromatograph (GC). Injection volumes for GCs can be 1 microliter (µL) or less and are difficult to reproduce. Differences in the volume of injected analyte leads to differences in the areas of the peaks in the chromatogram and any quantitative results are suspect.

To compensate for this error, a known amount of an internal standard (a second compound that does not interfere with the analysis of the primary analyte) is added to all solutions (standards and unknowns). This way if the injection volumes (and hence the peak areas) differ slightly, the ratio of the areas of the analyte and the internal standard will remain constant from one run to the next.

This comparison of runs also applies to solutions with different concentrations of the analyte. The area of the internal standard becomes the value to which all other areas are referenced. Below is the mathematical derivation and application of this method.

Consider an analysis of octane (C8H18) using nonane (C9H20) as the internal standard. The 3 chromatograms below are for 3 different samples.

The amount of octane in each sample is different, but the amount of nonane is the same (in practice this is not a requirement). Due to scaling, the areas of the nonane peak appear to have different areas, but in reality the areas are identical. Therefore, the relative amounts of octane in each sample increases in the order of mixture 1 (least) < mixture 3 < mixture 2 (most).

This conclusion is reached because the ratio of the area of octane to that of nonane is the least in mixture 1 and the most, in mixture 2. Mixture 3 has an intermediate ratio. This ratio can be written as $Area_{octane}/Area_{nonane}$ .

In chromatography, the area of a peak is proportional to the number of moles (n) times some constant of proportionality (k), Area = k×n. The number of moles of compound is equal to the concentration (molarity, M) times the volume, n = MV. From these equations, the following derivation is made:

${{Area_{octane}} \over {Area_{nonane}}}={{k_{octane}\times M_{octane}\times V_{octane}} \over {k_{nonane}\times M_{nonane}\times V_{nonane}}}$ Since both compounds are in the same solution and are injected together, the volume terms are equal and cancel out. The above equation is then rearranged to solve for the ratio of the k's. This ratio is then called the response factor, F.

$F={k_{octane} \over k_{nonane}}={{Area_{octane}/M_{octane}} \over {Area_{nonane}/M_{nonane}}}$ The response factor, F, is equal to the ratios of the k's, which are constant. Therefore, F is constant. What this means is that regardless of the amounts of octane and nonane in solution, the ratio of the ratios of area to concentration will always yield a constant.

In practice, a solution containing known amounts of both octane and nonane is injected into a GC and a response factor, F, is calculated. Then a separate solution with an unknown amount of octane and a known amount of nonane is injected. The response factor is applied to the data from the second solution and the unknown concentration of the octane is found.

${\left({{Area_{octane}/M_{octane}} \over {Area_{nonane}/M_{nonane}}}\right)}_{1}=F={\left({{Area_{octane}/M_{octane}} \over {Area_{nonane}/M_{nonane}}}\right)}_{2}$ This example deals with the analysis of octane and nonane, but can be applied to any two compounds.

## Related Research Articles Analytical chemistry studies and uses instruments and methods used to separate, identify, and quantify matter. In practice, separation, identification or quantification may constitute the entire analysis or be combined with another method. Separation isolates analytes. Qualitative analysis identifies analytes, while quantitative analysis determines the numerical amount or concentration.

In chemistry, concentration is the abundance of a constituent divided by the total volume of a mixture. Several types of mathematical description can be distinguished: mass concentration, molar concentration, number concentration, and volume concentration. A concentration can be any kind of chemical mixture, but most frequently solutes and solvents in solutions. The molar (amount) concentration has variants such as normal concentration and osmotic concentration.

In thermodynamics, the specific heat capacity of a substance is the heat capacity of a sample of the substance divided by the mass of the sample. Informally, it is the amount of energy that must be added, in the form of heat, to one unit of mass of the substance in order to cause an increase of one unit in temperature. The SI unit of specific heat is joule per kelvin and kilogram, J/(K kg). For example, at a temperature of 25 °C, the heat required to raise the temperature of 1 kg of water by 1 K is 4179.6 joules, meaning that the specific heat capacity of water is 4179.6 J·kg−1·K−1. Titration is a common laboratory method of quantitative chemical analysis to determine the concentration of an identified analyte. A reagent, termed the titrant or titrator, is prepared as a standard solution of known concentration and volume. The titrant reacts with a solution of analyte to determine the analyte's concentration. The volume of titrant that reacted with the analyte is termed the titration volume.

The gas constant is denoted by the symbol R or R. It is equivalent to the Boltzmann constant, but expressed in units of energy per temperature increment per mole, i.e. the pressure–volume product, rather than energy per temperature increment per particle. The constant is also a combination of the constants from Boyle's law, Charles's law, Avogadro's law, and Gay-Lussac's law. It is a physical constant that is featured in many fundamental equations in the physical sciences, such as the ideal gas law, the Arrhenius equation, and the Nernst equation. X-ray photoelectron spectroscopy (XPS) is a surface-sensitive quantitative spectroscopic technique based on the photoelectric effect that can identify the elements that exist within a material or are covering its surface, as well as their chemical state, and the overall electronic structure and density of the electronic states in the material. XPS is a powerful measurement technique because it not only shows what elements are present, but also what other elements they are bonded to. The technique can be used in line profiling of the elemental composition across the surface, or in depth profiling when paired with ion-beam etching. It is often applied to study chemical processes in the materials in their as-received state or after cleavage, scraping, exposure to heat, reactive gasses or solutions, ultraviolet light, or during ion implantation. Ultraviolet–visible spectroscopy or ultraviolet–visible spectrophotometry refers to absorption spectroscopy or reflectance spectroscopy in part of the ultraviolet and the full, adjacent visible spectral regions. This means it uses light in the visible and adjacent ranges. The absorption or reflectance in the visible range directly affects the perceived color of the chemicals involved. In this region of the electromagnetic spectrum, atoms and molecules undergo electronic transitions. Absorption spectroscopy is complementary to fluorescence spectroscopy, in that fluorescence deals with transitions from the excited state to the ground state, while absorption measures transitions from the ground state to the excited state. High-performance liquid chromatography (HPLC), formerly referred to as high-pressure liquid chromatography, is a technique in analytical chemistry used to separate, identify, and quantify each component in a mixture. It relies on pumps to pass a pressurized liquid solvent containing the sample mixture through a column filled with a solid adsorbent material. Each component in the sample interacts slightly differently with the adsorbent material, causing different flow rates for the different components and leading to the separation of the components as they flow out of the column.

In physical chemistry, Henry's law is a gas law that states that the amount of dissolved gas in a liquid is proportional to its partial pressure above the liquid. The proportionality factor is called Henry's law constant. It was formulated by the English chemist William Henry, who studied the topic in the early 19th century. In his publication about the quantity of gases absorbed by water, he described the results of his experiments:

… water takes up, of gas condensed by one, two, or more additional atmospheres, a quantity which, ordinarily compressed, would be equal to twice, thrice, &c. the volume absorbed under the common pressure of the atmosphere. In analytical chemistry, a calibration curve, also known as a standard curve, is a general method for determining the concentration of a substance in an unknown sample by comparing the unknown to a set of standard samples of known concentration. A calibration curve is one approach to the problem of instrument calibration; other standard approaches may mix the standard into the unknown, giving an internal standard. Gas chromatography (GC) is a common type of chromatography used in analytical chemistry for separating and analyzing compounds that can be vaporized without decomposition. Typical uses of GC include testing the purity of a particular substance, or separating the different components of a mixture. In some situations, GC may help in identifying a compound. In preparative chromatography, GC can be used to prepare pure compounds from a mixture.

Cavity ring-down spectroscopy (CRDS) is a highly sensitive optical spectroscopic technique that enables measurement of absolute optical extinction by samples that scatter and absorb light. It has been widely used to study gaseous samples which absorb light at specific wavelengths, and in turn to determine mole fractions down to the parts per trillion level. The technique is also known as cavity ring-down laser absorption spectroscopy (CRLAS).

Micellar electrokinetic chromatography (MEKC) is a chromatography technique used in analytical chemistry. It is a modification of capillary electrophoresis (CE), extending its functionality to neutral analytes, where the samples are separated by differential partitioning between micelles and a surrounding aqueous buffer solution. Isotope dilution analysis is a method of determining the quantity of chemical substances. In its most simple conception, the method of isotope dilution comprises the addition of known amounts of isotopically-enriched substance to the analyzed sample. Mixing of the isotopic standard with the sample effectively "dilutes" the isotopic enrichment of the standard and this forms the basis for the isotope dilution method. Isotope dilution is classified as a method of internal standardisation, because the standard is added directly to the sample. In addition, unlike traditional analytical methods which rely on signal intensity, isotope dilution employs signal ratios. Owing to both of these advantages, the method of isotope dilution is regarded among chemistry measurement methods of the highest metrological standing. The method of standard addition is a type of quantitative analysis approach often used in analytical chemistry whereby the standard is added directly to the aliquots of analyzed sample. This method is used in situations where sample matrix also contributes to the analytical signal, a situation known as the matrix effect, thus making it impossible to compare the analytical signal between sample and standard using the traditional calibration curve approach.

A total ionic strength adjustment buffer (TISAB) is a buffer solution which increases the ionic strength of a solution to a relatively high level. This is important for potentiometric measurements, including ion selective electrodes, because they measure the activity of the analyte rather than its concentration. TISAB essentially masks minor changes made in the ionic strength of the solution and hence increases the accuracy of the reading.

In chemical analysis, matrix refers to the components of a sample other than the analyte of interest. The matrix can have a considerable effect on the way the analysis is conducted and the quality of the results are obtained; such effects are called matrix effects. For example, the ionic strength of the solution can have an effect on the activity coefficients of the analytes. The most common approach for accounting for matrix effects is to build a calibration curve using standard samples with known analyte concentration and which try to approximate the matrix of the sample as much as possible. This is especially important for solid samples where there is a strong matrix influence. In cases with complex or unknown matrices, the standard addition method can be used. In this technique, the response of the sample is measured and recorded, for example, using an electrode selective for the analyte. Then, a small volume of standard solution is added and the response is measured again. Ideally, the standard addition should increase the analyte concentration by a factor of 1.5 to 3, and several additions should be averaged. The volume of standard solution should be small enough to disturb the matrix as little as possible.

Binding selectivity is defined with respect to the binding of ligands to a substrate forming a complex. Binding selectivity describes how a ligand may bind more preferentially to one receptor than another. A selectivity coefficient is the equilibrium constant for the reaction of displacement by one ligand of another ligand in a complex with the substrate. Binding selectivity is of major importance in biochemistry and in chemical separation processes.

Equilibrium chemistry is concerned with systems in chemical equilibrium. The unifying principle is that the free energy of a system at equilibrium is the minimum possible, so that the slope of the free energy with respect to the reaction coordinate is zero. This principle, applied to mixtures at equilibrium provides a definition of an equilibrium constant. Applications include acid–base, host–guest, metal–complex, solubility, partition, chromatography and redox equilibria.

The Polyarc reactor is a scientific instrument for the measurement of organic molecules. The reactor is paired with a flame ionization detector (FID) in a gas chromatograph (GC) to improve the sensitivity of the FID and give a uniform detector response for all organic molecules (GC-Polyarc/FID).

1. Ramus TL; Hein SJ; Thomas LC (August 1987). "Determinations of polychlorinated biphenyl isomers by response factor calibration". J. Chromatogr. 404 (1): 155–62. doi:10.1016/S0021-9673(01)86846-1. PMID   3119645.
2. Orange Book. Compendium of Analytical Nomenclature (PDF). IUPAC.