Photomultiplier tubes (photomultipliers or PMTs for short), members of the class of vacuum tubes, and more specifically vacuum phototubes, are extremely sensitive detectors of light in the ultraviolet, visible, and near-infrared ranges of the electromagnetic spectrum. These detectors multiply the current produced by incident light by as much as 100 million times or 108 (i.e., 160 dB), in multiple dynode stages, enabling (for example) individual photons to be detected when the incident flux of light is low.
In electronics, a vacuum tube, an electron tube, or valve or, colloquially, a tube, is a device that controls electric current flow in a high vacuum between electrodes to which an electric potential difference has been applied.
A phototube or photoelectric cell is a type of gas-filled or vacuum tube that is sensitive to light. Such a tube is more correctly called a 'photoemissive cell' to distinguish it from photovoltaic or photoconductive cells. Phototubes were previously more widely used but are now replaced in many applications by solid state photodetectors. The photomultiplier tube is one of the most sensitive light detectors, and is still widely used in physics research.
Ultraviolet (UV) designates a band of the electromagnetic spectrum with wavelength from 10 nm to 400 nm, shorter than that of visible light but longer than X-rays. UV radiation is present in sunlight, and contributes about 10% of the total light output of the Sun. It is also produced by electric arcs and specialized lights, such as mercury-vapor lamps, tanning lamps, and black lights. Although long-wavelength ultraviolet is not considered an ionizing radiation because its photons lack the energy to ionize atoms, it can cause chemical reactions and causes many substances to glow or fluoresce. Consequently, the chemical and biological effects of UV are greater than simple heating effects, and many practical applications of UV radiation derive from its interactions with organic molecules.
The combination of high gain, low noise, high frequency response or, equivalently, ultra-fast response, and large area of collection has maintained photomultipliers an essential place in low light level spectroscopy, confocal microscopy, Raman spectroscopy, fluorescence spectroscopy, nuclear and particle physics, astronomy, medical diagnostics including blood tests, medical imaging, motion picture film scanning (telecine), radar jamming, and high-end image scanners known as drum scanners. Elements of photomultiplier technology, when integrated differently, are the basis of night vision devices. Research that analyzes light scattering, such as the study of polymers in solution, often uses a laser and a PMT to collect the scattered light data.
In electronics, gain is a measure of the ability of a two-port circuit to increase the power or amplitude of a signal from the input to the output port by adding energy converted from some power supply to the signal. It is usually defined as the mean ratio of the signal amplitude or power at the output port to the amplitude or power at the input port. It is often expressed using the logarithmic decibel (dB) units. A gain greater than one, that is amplification, is the defining property of an active component or circuit, while a passive circuit will have a gain of less than one.
In electronics, noise is an unwanted disturbance in an electrical signal. Noise generated by electronic devices varies greatly as it is produced by several different effects.
Spectroscopy is the study of the interaction between matter and electromagnetic radiation. Historically, spectroscopy originated through the study of visible light dispersed according to its wavelength, by a prism. Later the concept was expanded greatly to include any interaction with radiative energy as a function of its wavelength or frequency, predominantly in the electromagnetic spectrum, though matter waves and acoustic waves can also be considered forms of radiative energy; recently, with tremendous difficulty, even gravitational waves have been associated with a spectral signature in the context of LIGO and laser interferometry. Spectroscopic data are often represented by an emission spectrum, a plot of the response of interest as a function of wavelength or frequency.
Semiconductor devices, particularly avalanche photodiodes, are alternatives to photomultipliers; however, photomultipliers are uniquely well-suited for applications requiring low-noise, high-sensitivity detection of light that is imperfectly collimated.
Semiconductor devices are electronic components that exploit the electronic properties of semiconductor material, principally silicon, germanium, and gallium arsenide, as well as organic semiconductors. Semiconductor devices have replaced thermionic devices in most applications. They use electronic conduction in the solid state as opposed to the gaseous state or thermionic emission in a high vacuum.
An avalanche photodiode (APD) is a highly sensitive semiconductor electronic device that exploits the photoelectric effect to convert light to electricity. APDs can be thought of as photodetectors that provide a built-in first stage of gain through avalanche multiplication. From a functional standpoint, they can be regarded as the semiconductor analog of photomultipliers. By applying a high reverse bias voltage, APDs show an internal current gain effect due to impact ionization. However, some silicon APDs employ alternative doping and beveling techniques compared to traditional APDs that allow greater voltage to be applied before breakdown is reached and hence a greater operating gain. In general, the higher the reverse voltage, the higher the gain. Among the various expressions for the APD multiplication factor (M), an instructive expression is given by the formula
Photomultipliers are typically constructed with an evacuated glass housing (using an extremely tight and durable glass-to-metal seal like other vacuum tubes), containing a photocathode, several dynodes, and an anode. Incident photons strike the photocathode material, which is usually a thin vapor-deposited conducting layer on the inside of the entry window of the device. Electrons are ejected from the surface as a consequence of the photoelectric effect. These electrons are directed by the focusing electrode toward the electron multiplier, where electrons are multiplied by the process of secondary emission.
Glass-to-metal seals are a very important element of the construction of vacuum tubes, electric discharge tubes, incandescent light bulbs, glass encapsulated semiconductor diodes, reed switches, pressure tight glass windows in metal cases, and metal or ceramic packages of electronic components.
A photocathode is a negatively charged electrode in a light detection device such as a photomultiplier or phototube that is coated with a photosensitive compound. When this is struck by a quantum of light (photon), the absorbed energy causes electron emission due to the photoelectric effect.
A dynode is an electrode in a vacuum tube that serves as an electron multiplier through secondary emission. The first tube to incorporate a dynode was the dynatron, an ancestor of the magnetron, which used a single dynode. Photomultiplier and video camera tubes generally include a series of dynodes, each at a more positive electrical potential than its predecessor. Secondary emission occurs at the surface of each dynode. Such an arrangement is able to amplify the tiny current emitted by the photocathode, typically by a factor of one million.
The electron multiplier consists of a number of electrodes called dynodes. Each dynode is held at a more positive potential, by ≈100 Volts, than the preceding one. A primary electron leaves the photocathode with the energy of the incoming photon, or about 3 eV for "blue" photons, minus the work function of the photocathode. A small group of primary electrons is created by the arrival of a group of initial photons. (In Fig. 1, the number of primary electrons in the initial group is proportional to the energy of the incident high energy gamma ray.) The primary electrons move toward the first dynode because they are accelerated by the electric field. They each arrive with ≈100 eV kinetic energy imparted by the potential difference. Upon striking the first dynode, more low energy electrons are emitted, and these electrons are in turn accelerated toward the second dynode. The geometry of the dynode chain is such that a cascade occurs with an exponentially-increasing number of electrons being produced at each stage. For example, if at each stage an average of 5 new electrons are produced for each incoming electron, and if there are 12 dynode stages, then at the last stage one expects for each primary electron about 512 ≈ 108 electrons. This last stage is called the anode. This large number of electrons reaching the anode results in a sharp current pulse that is easily detectable, for example on an oscilloscope, signaling the arrival of the photon(s) at the photocathode ≈50 nanoseconds earlier.
In solid-state physics, the work function is the minimum thermodynamic work needed to remove an electron from a solid to a point in the vacuum immediately outside the solid surface. Here "immediately" means that the final electron position is far from the surface on the atomic scale, but still too close to the solid to be influenced by ambient electric fields in the vacuum. The work function is not a characteristic of a bulk material, but rather a property of the surface of the material.
An anode is an electrode through which the conventional current enters into a polarized electrical device. This contrasts with a cathode, an electrode through which conventional current leaves an electrical device. A common mnemonic is ACID for "anode current into device". The direction of conventional current in a circuit is opposite to the direction of electron flow, so electrons flow out the anode into the outside circuit. In a galvanic cell, the anode is the electrode at which the oxidation reaction occurs.
The necessary distribution of voltage along the series of dynodes is created by a voltage divider chain, as illustrated in Fig. 2. In the example, the photocathode is held at a negative high voltage of order 1000 V, while the anode is very close to ground potential. The capacitors across the final few dynodes act as local reservoirs of charge to help maintain the voltage on the dynodes while electron avalanches propagate through the tube. Many variations of design are used in practice; the design shown is merely illustrative.
There are two common photomultiplier orientations, the head-on or end-on (transmission mode) design, as shown above, where light enters the flat, circular top of the tube and passes the photocathode, and the side-on design (reflection mode), where light enters at a particular spot on the side of the tube, and impacts on an opaque photocathode. The side-on design is used, for instance, in the type 931, the first mass-produced PMT. Besides the different photocathode materials, performance is also affected by the transmission of the window material that the light passes through, and by the arrangement of the dynodes. Many photomultiplier models are available having various combinations of these, and other, design variables. The manufacturers manuals provide the information needed to choose an appropriate design for a particular application.
The invention of the photomultiplier is predicated upon two prior achievements, the separate discoveries of the photoelectric effect and of secondary emission.
The first demonstration of the photoelectric effect was carried out in 1887 by Heinrich Hertz using ultraviolet light.Significant for practical applications, Elster and Geitel two years later demonstrated the same effect using visible light striking alkali metals (potassium and sodium). The addition of caesium, another alkali metal, has permitted the range of sensitive wavelengths to be extended towards longer wavelengths in the red portion of the visible spectrum.
Historically, the photoelectric effect is associated with Albert Einstein, who relied upon the phenomenon to establish the fundamental principle of quantum mechanics in 1905,an accomplishment for which Einstein received the 1921 Nobel Prize. It is worthwhile to note that Heinrich Hertz, working 18 years earlier, had not recognized that the kinetic energy of the emitted electrons is proportional to the frequency but independent of the optical intensity. This fact implied a discrete nature of light, i.e. the existence of quanta, for the first time.
The phenomenon of secondary emission (the ability of electrons in a vacuum tube to cause the emission of additional electrons by striking an electrode) was, at first, limited to purely electronic phenomena and devices (which lacked photosensitivity). In 1899 the effect was first reported by Villard.In 1902, Austin and Starke reported that the metal surfaces impacted by electron beams emitted a larger number of electrons than were incident. The application of the newly discovered secondary emission to the amplification of signals was only proposed after World War I by Westinghouse scientist Joseph Slepian in a 1919 patent.
The ingredients for inventing the photomultiplier were coming together during the 1920s as the pace of vacuum tube technology accelerated. The primary goal for many, if not most, workers was the need for a practical television camera technology. Television had been pursued with primitive prototypes for decades prior to the 1934 introduction of the first practical camera (the iconoscope). Early prototype television cameras lacked sensitivity. Photomultiplier technology was pursued to enable television camera tubes, such as the iconoscope and (later) the orthicon, to be sensitive enough to be practical. So the stage was set to combine the dual phenomena of photoemission (i.e., the photoelectric effect) with secondary emission, both of which had already been studied and adequately understood, to create a practical photomultiplier.
The first documented photomultiplier demonstration dates to the early 1934 accomplishments of an RCA group based in Harrison, NJ. Harley Iams and Bernard Salzberg were the first to integrate a photoelectric-effect cathode and single secondary emission amplification stage in a single vacuum envelope and the first to characterize its performance as a photomultiplier with electron amplification gain. These accomplishments were finalized prior to June 1934 as detailed in the manuscript submitted to Proceedings of the Institute of Radio Engineers (Proc. IRE). kHz.The device consisted of a semi-cylindrical photocathode, a secondary emitter mounted on the axis, and a collector grid surrounding the secondary emitter. The tube had a gain of about eight and operated at frequencies well above 10
Higher gains were sought than those available from the early single-stage photomultipliers. However, it is an empirical fact that the yield of secondary electrons is limited in any given secondary emission process, regardless of acceleration voltage. Thus, any single-stage photomultiplier is limited in gain. At the time the maximum first-stage gain that could be achieved was approximately 10 (very significant developments in the 1960s permitted gains above 25 to be reached using negative electron affinity dynodes). For this reason, multiple-stage photomultipliers, in which the photoelectron yield could be multiplied successively in several stages, were an important goal. The challenge was to cause the photoelectrons to impinge on successively higher-voltage electrodes rather than to travel directly to the highest voltage electrode. Initially this challenge was overcome by using strong magnetic fields to bend the electrons' trajectories. Such a scheme had earlier been conceived by inventor J. Slepian by 1919 (see above). Accordingly, leading international research organizations turned their attention towards improving photomultiplers to achieve higher gain with multiple stages.
In the USSR, RCA-manufactured radio equipment was introduced on a large scale by Joseph Stalin to construct broadcast networks, and the newly formed All-Union Scientific Research Institute for Television was gearing up a research program in vacuum tubes that was advanced for its time and place. Numerous visits were made by RCA scientific personnel to the USSR in the 1930s, prior to the Cold War, to instruct the Soviet customers on the capabilities of RCA equipment and to investigate customer needs.During one of these visits, in September 1934, RCA's Vladimir Zworykin was shown the first multiple-dynode photomultiplier, or photoelectron multiplier. This pioneering device was proposed by Leonid A. Kubetsky in 1930 which he subsequently built in 1934. The device achieved gains of 1000x or more when demonstrated in June 1934. The work was submitted for print publication only two years later, in July 1936 as emphasized in a recent 2006 publication of the Russian Academy of Sciences (RAS), which terms it "Kubetsky's Tube." The Soviet device used a magnetic field to confine the secondary electrons and relied on the Ag-O-Cs photocathode which had been demonstrated by General Electric in the 1920s.
By October 1935, Vladimir Zworykin, George Ashmun Morton, and Louis Malter of RCA in Camden, NJ submitted their manuscript describing the first comprehensive experimental and theoretical analysis of a multiple dynode tube — the device later called a photomultiplier— to Proc. IRE. The RCA prototype photomultipliers also used an Ag-O-Cs (silver oxide-caesium) photocathode. They exhibited a peak quantum efficiency of 0.4% at 800 nm.
Whereas these early photomultipliers used the magnetic field principle, electrostatic photomultipliers (with no magnetic field) were demonstrated by Jan Rajchman of RCA Laboratories in Princeton, NJ in the late 1930s and became the standard for all future commercial photomultipliers. The first mass-produced photomultiplier, the Type 931, was of this design and is still commercially produced today.
Also in 1936, a much improved photocathode, Cs3Sb (caesium-antimony), was reported by P. Görlich. nm, and was used in the first commercially successful photomultipliers manufactured by RCA (i.e., the 931-type) both as a photocathode and as a secondary-emitting material for the dynodes. Different photocathodes provided differing spectral responses.The caesium-antimony photocathode had a dramatically improved quantum efficiency of 12% at 400
In the early 1940s, the JEDEC (Joint Electron Device Engineering Council), an industry committee on standardization, developed a system of designating spectral responses.The philosophy included the idea that the product's user need only be concerned about the response of the device rather than how the device may be fabricated. Various combinations of photocathode and window materials were assigned "S-numbers" (spectral numbers) ranging from S-1 through S-40, which are still in use today. For example, S-11 uses the caesium-antimony photocathode with a lime glass window, S-13 uses the same photocathode with a fused silica window, and S-25 uses a so-called "multialkali" photocathode (Na-K-Sb-Cs, or sodium-potassium-antimony-caesium) that provides extended response in the red portion of the visible light spectrum. No suitable photoemissive surfaces have yet been reported to detect wavelengths longer than approximately 1700 nanometers, which can be approached by a special (InP/InGaAs(Cs)) photocathode.
For decades, RCA was responsible for performing the most important work in developing and refining photomultipliers. RCA was also largely responsible for the commercialization of photomultiplers. The company compiled and published an authoritative and widely used Photomultiplier Handbook.RCA provided printed copies free upon request. The handbook, which continues to be made available online at no cost by the successors to RCA, is considered to be an essential reference.
Following a corporate break-up in the late 1980s involving the acquisition of RCA by General Electric and disposition of the divisions of RCA to numerous third parties, RCA's photomultiplier business became an independent company.
The Lancaster, Pennsylvania facility was opened by the U.S. Navy in 1942 and operated by RCA for the manufacture of radio and microwave tubes. Following World War II, the naval facility was acquired by RCA. RCA Lancaster, as it became known, was the base for the development and the production of commercial television products. In subsequent years other products were added, such as "cathode-ray" tubes, photomultiplier tubes, motion-sensing light control switches, and closed-circuit television systems.
Burle Industries, as a successor to the RCA Corporation, carried the RCA photomultiplier business forward after 1986, based in the Lancaster, Pennsylvania facility. The 1986 acquisition of RCA by General Electric resulted in the divestiture of the RCA Lancaster New Products Division. Hence, 45 years after being founded by the U.S. Navy, its management team, led by Erich Burlefinger, purchased the division and in 1987 founded Burle Industries.
In 2005, after eighteen years as an independent enterprise, Burle Industries and a key subsidiary were acquired by Photonis, a European holding company Photonis Group. Following the acquisition, Photonis was composed of Photonis Netherlands, Photonis France, Photonis USA, and Burle Industries. Photonis USA operates the former Galileo Corporation Scientific Detector Products Group (Sturbridge, Massachusetts), which had been purchased by Burle Industries in 1999. The group is known for microchannel plate detector (MCP) electron multipliers—an integrated micro-vacuum tube version of photomultipliers. MCPs are used for imaging and scientific applications, including night vision devices.
On 9 March 2009, Photonis announced that it would cease all production of photomultipliers at both the Lancaster, Pennsylvania and the Brive, France plants.
The Japan-based company Hamamatsu Photonics (also known as Hamamatsu) has emerged since the 1950s as a leader in the photomultiplier industry. Hamamatsu, in the tradition of RCA, has published its own handbook, which is available without cost on the company's website.Hamamatsu uses different designations for particular photocathode formulations and introduces modifications to these designations based on Hamamatsu's proprietary research and development.
The photocathodes can be made of a variety of materials, with different properties. Typically the materials have low work function and are therefore prone to thermionic emission, causing noise and dark current, especially the materials sensitive in infrared; cooling the photocathode lowers this thermal noise. The most common photocathode materials are nm. High dark current; used mainly in near-infrared, with the photocathode cooled; GaAs:Cs, caesium-activated gallium arsenide, flat response from 300 to 850 nm, fading towards ultraviolet and to 930 nm; InGaAs:Cs, caesium-activated indium gallium arsenide, higher infrared sensitivity than GaAs:Cs, between 900–1000 nm much higher signal-to-noise ratio than Ag-O-Cs; Sb-Cs, (also called S11) caesium-activated antimony, used for reflective mode photocathodes; response range from ultraviolet to visible, widely used; bialkali (Sb-K-Cs, Sb-Rb-Cs), caesium-activated antimony-rubidium or antimony-potassium alloy, similar to Sb:Cs, with higher sensitivity and lower noise. can be used for transmission-mode; favorable response to a NaI:Tl scintillator flashes makes them widely used in gamma spectroscopy and radiation detection; high-temperature bialkali (Na-K-Sb), can operate up to 175 °C, used in well logging, low dark current at room temperature; multialkali (Na-K-Sb-Cs), (also called S20), wide spectral response from ultraviolet to near-infrared, special cathode processing can extend range to 930 nm, used in broadband spectrophotometers; solar-blind (Cs-Te, Cs-I), sensitive to vacuum-UV and ultraviolet, insensitive to visible light and infrared (Cs-Te has cutoff at 320 nm, Cs-I at 200 nm).Ag-O-Cs (also called S1) transmission-mode, sensitive from 300–1200
The windows of the photomultipliers act as wavelength filters; this may be irrelevant if the cutoff wavelengths are outside of the application range or outside of the photocathode sensitivity range, but special care has to be taken for uncommon wavelengths. Borosilicate glass is commonly used for near-infrared to about 300 nm. High borate borosilicate glasses exist also in high UV transmission versions with high transmission also at 254 nm. Glass with very low content of potassium can be used with bialkali photocathodes to lower the background radiation from the potassium-40 isotope. Ultraviolet glass transmits visible and ultraviolet down to 185 nm. Used in spectroscopy. Synthetic silica transmits down to 160 nm, absorbs less UV than fused silica. Different thermal expansion than kovar (and than borosilicate glass that's expansion-matched to kovar), a graded seal needed between the window and the rest of the tube. The seal is vulnerable to mechanical shocks. Magnesium fluoride transmits ultraviolet down to 115 nm. Hygroscopic, though less than other alkali halides usable for UV windows.
Photomultiplier tubes typically utilize 1000 to 2000 volts to accelerate electrons within the chain of dynodes. (See Figure near top of article.) The most negative voltage is connected to the cathode, and the most positive voltage is connected to the anode. Negative high-voltage supplies (with the positive terminal grounded) are often preferred, because this configuration enables the photocurrent to be measured at the low voltage side of the circuit for amplification by subsequent electronic circuits operating at low voltage. However, with the photocathode at high voltage, leakage currents sometimes result in unwanted "dark current" pulses that may affect the operation. Voltages are distributed to the dynodes by a resistive voltage divider, although variations such as active designs (with transistors or diodes) are possible. The divider design, which influences frequency response or rise time, can be selected to suit varying applications. Some instruments that use photomultipliers have provisions to vary the anode voltage to control the gain of the system.
While powered (energized), photomultipliers must be shielded from ambient light to prevent their destruction through overexcitation. In some applications this protection is accomplished mechanically by electrical interlocks or shutters that protect the tube when the photomultiplier compartment is opened. Another option is to add overcurrent protection in the external circuit, so that when the measured anode current exceeds a safe limit, the high voltage is reduced.
If used in a location with strong magnetic fields, which can curve electron paths, steer the electrons away from the dynodes and cause loss of gain, photomultipliers are usually magnetically shielded by a layer of soft iron or mu-metal. This magnetic shield is often maintained at cathode potential. When this is the case, the external shield must also be electrically insulated because of the high voltage on it. Photomultipliers with large distances between the photocathode and the first dynode are especially sensitive to magnetic fields.
Photomultipliers were the first electric eye devices, being used to measure interruptions in beams of light. Photomultipliers are used in conjunction with scintillators to detect Ionizing radiation by means of hand held and fixed radiation protection instruments, and particle radiation in physics experiments.Photomultipliers are used in research laboratories to measure the intensity and spectrum of light-emitting materials such as compound semiconductors and quantum dots. Photomultipliers are used as the detector in many spectrophotometers. This allows an instrument design that escapes the thermal noise limit on sensitivity, and which can therefore substantially increase the dynamic range of the instrument.
Photomultipliers are used in numerous medical equipment designs. For example, blood analysis devices used by clinical medical laboratories, such as flow cytometers, utilize photomultipliers to determine the relative concentration of various components in blood samples, in combination with optical filters and incandescent lamps. An array of photomultipliers is used in a gamma camera. Photomultipliers are typically used as the detectors in flying-spot scanners.
After 50 years, during which solid-state electronic components have largely displaced the vacuum tube, the photomultiplier remains a unique and important optoelectronic component. Perhaps its most useful quality is that it acts, electronically, as a nearly perfect current source, owing to the high voltage utilized in extracting the tiny currents associated with weak light signals. There is no Johnson noise associated with photomultiplier signal currents, even though they are greatly amplified, e.g., by 100 thousand times (i.e., 100 dB) or more. The photocurrent still contains shot noise.
Photomultiplier-amplified photocurrents can be electronically amplified by a high-input-impedance electronic amplifier (in the signal path subsequent to the photomultiplier), thus producing appreciable voltages even for nearly infinitesimally small photon fluxes. Photomultipliers offer the best possible opportunity to exceed the Johnson noise for many configurations. The aforementioned refers to measurement of light fluxes that, while small, nonetheless amount to a continuous stream of multiple photons.
For smaller photon fluxes, the photomultiplier can be operated in photon-counting, or Geiger, mode (see also Single-photon avalanche diode). In Geiger mode the photomultiplier gain is set so high (using high voltage) that a single photo-electron resulting from a single photon incident on the primary surface generates a very large current at the output circuit. However, owing to the avalanche of current, a reset of the photomultiplier is required. In either case, the photomultiplier can detect individual photons. The drawback, however, is that not every photon incident on the primary surface is counted either because of less-than-perfect efficiency of the photomultiplier, or because a second photon can arrive at the photomultiplier during the "dead time" associated with a first photon and never be noticed.
A photomultiplier will produce a small current even without incident photons; this is called the dark current. Photon-counting applications generally demand photomultipliers designed to minimise dark current.
Nonetheless, the ability to detect single photons striking the primary photosensitive surface itself reveals the quantization principle that Einstein put forth. Photon counting (as it is called) reveals that light, not only being a wave, consists of discrete particles (i.e., photons).
A cathode is the electrode from which a conventional current leaves a polarized electrical device. This definition can be recalled by using the mnemonic CCD for Cathode Current Departs. A conventional current describes the direction in which positive charges move. Electrons have a negative electrical charge, so the movement of electrons is opposite to that of the conventional current flow. Consequently, the mnemonic cathode current departs also means that electrons flow into the device's cathode from the external circuit.
The photoelectric effect is the emission of electrons or other free carriers when light falls on a material. Electrons emitted in this manner can be called photo electrons. This phenomenon is commonly studied in electronic physics, as well as in fields of chemistry, such as quantum chemistry or electrochemistry.
A photodiode is a semiconductor device that converts light into an electrical current. The current is generated when photons are absorbed in the photodiode. Photodiodes may contain optical filters, built-in lenses, and may have large or small surface areas. Photodiodes usually have a slower response time as their surface area increases. The common, traditional solar cell used to generate electric solar power is a large area photodiode.
A tetrode is a vacuum tube having four active electrodes. The four electrodes in order from the centre are: a thermionic cathode, first and second grids and a plate. There are several varieties of tetrodes, the most common being the screen-grid tube and the beam tetrode. In screen-grid tubes and beam tetrodes, the first grid is the control grid and the second grid is the screen grid. In other tetrodes one of the grids is a control grid, while the other may have a variety of functions.
A cold cathode is a cathode that is not electrically heated by a filament. A cathode may be considered "cold" if it emits more electrons than can be supplied by thermionic emission alone. It is used in gas-discharge lamps, such as neon lamps, discharge tubes, and some types of vacuum tube. The other type of cathode is a hot cathode, which is heated by electric current passing through a filament. A cold cathode does not necessarily operate at a low temperature: it is often heated to its operating temperature by other methods, such as the current passing from the cathode into the gas.
A scintillation counter is an instrument for detecting and measuring ionizing radiation by using the excitation effect of incident radiation on a scintillating material, and detecting the resultant light pulses.
The Franck–Hertz experiment was the first electrical measurement to clearly show the quantum nature of atoms, and thus "transformed our understanding of the world". It was presented on April 24, 1914, to the German Physical Society in a paper by James Franck and Gustav Hertz. Franck and Hertz had designed a vacuum tube for studying energetic electrons that flew through a thin vapor of mercury atoms. They discovered that, when an electron collided with a mercury atom, it could lose only a specific quantity of its kinetic energy before flying away. This energy loss corresponds to decelerating the electron from a speed of about 1.3 million meters per second to zero. A faster electron does not decelerate completely after a collision, but loses precisely the same amount of its kinetic energy. Slower electrons merely bounce off mercury atoms without losing any significant speed or kinetic energy.
Secondary emission in physics is a phenomenon where primary incident particles of sufficient energy, when hitting a surface or passing through some material, induce the emission of secondary particles. The term often refers to the emission of electrons when charged particles like electrons or ions in a vacuum tube strike a metal surface; these are called secondary electrons. In this case, the number of secondary electrons emitted per incident particle is called secondary emission yield. If the secondary particles are ions, the effect is termed secondary ion emission. Secondary electron emission is used in photomultiplier tubes and image intensifier tubes to amplify the small number of photoelectrons produced by photoemission, making the tube more sensitive. It also occurs as an undesirable side effect in electronic vacuum tubes when electrons from the cathode strike the anode, and can cause parasitic oscillation.
An image intensifier or image intensifier tube is a vacuum tube device for increasing the intensity of available light in an optical system to allow use under low-light conditions, such as at night, to facilitate visual imaging of low-light processes, such as fluorescence of materials in X-rays or gamma rays, or for conversion of non-visible light sources, such as near-infrared or short wave infrared to visible. They operate by converting photons of light into electrons, amplifying the electrons, and then converting the amplified electrons back into photons for viewing. They are used in devices such as night vision goggles.
Photodetectors, also called photosensors, are sensors of light or other electromagnetic radiation. A photo detector has a p–n junction that converts light photons into current. The absorbed photons make electron–hole pairs in the depletion region. Photodiodes and photo transistors are a few examples of photo detectors. Solar cells convert some of the light energy absorbed into electrical energy.
A plate, usually called anode in Britain, is a type of electrode that forms part of a vacuum tube. It is usually made of sheet metal, connected to a wire which passes through the glass envelope of the tube to a terminal in the base of the tube, where it is connected to the external circuit. The plate is given a positive potential, and its function is to attract and capture the electrons emitted by the cathode. Although it is sometimes a flat plate, it is more often in the shape of a cylinder or flat open-ended box surrounding the other electrodes.
An electron multiplier is a vacuum-tube structure that multiplies incident charges. In a process called secondary emission, a single electron can, when bombarded on secondary-emissive material, induce emission of roughly 1 to 3 electrons. If an electric potential is applied between this metal plate and yet another, the emitted electrons will accelerate to the next metal plate and induce secondary emission of still more electrons. This can be repeated a number of times, resulting in a large shower of electrons all collected by a metal anode, all having been triggered by just one.
A pentode is an electronic device having five active electrodes. The term most commonly applies to a three-grid amplifying vacuum tube, which was invented by Gilles Holst and Bernhard D.H. Tellegen in 1926. The pentode consists of an evacuated glass envelope containing five electrodes in this order: a cathode heated by a filament, a control grid, a screen grid, a suppressor grid, and a plate (anode). The pentode was developed from the tetrode tube by the addition of a third grid, the suppressor grid. This served to prevent secondary emission electrons emitted by the plate from reaching the screen grid, which caused instability and parasitic oscillations in the tetrode. The pentode is closely related to the beam tetrode. Pentodes were widely used in industrial and consumer electronic equipment such as radios and televisions until the 1960s, when they were replaced by transistors. Their main use now is in high power industrial applications such as radio transmitters. The obsolete consumer tubes are still used in a few legacy and specialty vacuum tube audio devices.
The Everhart-Thornley Detector is a secondary electron and back-scattered electron detector used in scanning electron microscopes (SEMs). It is named after its designers, Thomas E. Everhart and Richard F. M. Thornley who in 1960 published their design to increase the efficiency of existing secondary electron detectors by adding a light pipe to carry the photon signal from the scintillator inside the evacuated specimen chamber of the SEM to the photomultiplier outside the chamber. Prior to this Everhart had improved a design for a secondary electron detection by Vladimir Zworykin and J. A. Rajchman by changing the electron multiplier to a photomultiplier. The Everhart-Thornley Detector with its lightguide and highly efficient photomultiplier is the most frequently used detector in SEMs.
A micro-channel plate (MCP) is a planar component used for detection of single particles and low intensity impinging radiation. It is closely related to an electron multiplier, as both intensify single particles or photons by the multiplication of electrons via secondary emission. However, because a microchannel plate detector has many separate channels, it can additionally provide spatial resolution.
Photek Limited is a specialist manufacturer and global supplier of vacuum based tubes and camera systems for photon detection. Photek manufacture image intensifiers, solar-blind detectors, photomultipliers, streak tubes and a range of associated electronics and camera systems. The company was founded in 1991 by Jon Howorth, Ralph Powell, Martin Ingle, Geoff Holt and Mehmet Madakbas.
|Wikimedia Commons has media related to photomultiplier .|